The green buttons are links to more pages. If your cursor changes into a pointer when over one, then that button is an active link.
You can click on it to go to it's page. Please note that most of the
buttons act the same as in the actual hand controller. You can go
through them as you would on the hand controller.
If this is your first time operating the mount select the Quick Start
option. All buttons do take you to the appropriate menu. There is
also a menu that can select which default mode the Gemini-2 starts up
You will see this menu if you have "Ask if possible" selected as the
default start up mode. That menu is under "Mount" then "Startup" in a
You have 4 choices for startup.
"Quick Start" - This selection will take you to
menu's to select your "Mount Type", Enter your "Location", then "Time,
Time Zone and Date" Then you will go to the "Main Screen" This also does
a "Cold Start".
"Cold Start" - A cold start wipes out all stored
modeling. You need to have your mount positioned at want is called CWD.
This is with the counter weights down, and the Declination pointed
towards Polaris in the Northern Hemisphere, and the South Celestial Pole in the Southern Hemisphere.
The processor in the Gemini-2 uses this approximant position to start
it's calculations from. It expects you to do an alignment, or
synchronize on a bright star. When you align or synchronize on the
bright start, then the processor refines the known position in the sky and also
repositions the limits correctly. It also updates the All modeling calculations will
start from this point in space.
"Warm Start" - This is basically the same as a
cold start, but does not wipe out any models built. It also remembers
all your setting. You still must start with the mount pointed to CWD
position as in a cold start. If you have models built, but have moved
your Right Ascension axis or Declination axis, but not the location of
the mount itself, then you can use this startup mode. Warm Start uses
the CWD position as an approximant starting position, but expects you to do either an
alignment, or synchronize on a bright star. When you align or
synchronize on the bright start, then the processors re-centers the
model, and positions the limits correctly.
"Warm Restart" - This mode also remembers your
modeling and all setting. You can only use this mode if, and only if you
have not moved both the Right Ascension axis or Declination axis and
also have not moved the mount in position. All calculations will
start from this point.